PALM is a super-resolution microscopy technique and PALM is short for "Photoactivated Localization Microscopy". The technique was invented by Eric Betzig along with his colleague Harald Hess and this work was recognised internationally with the award of the 2014 Nobel Prize for Chemistry. Betzig shared the award with Stefan Hell and William Moerner for their contributions to breaking through the diffraction barrier in fluorescence microscopy.
PALM microscopy uses photoactivatable fluorophores to resolve spatial details of tightly packed molecules. Once activated by lasers, fluorophores emit for a short period but eventually bleach. The laser stochastically activates fluorophores until all have emitted. Crucially by activating them randomly and in smaller numbers, a more accurate view of positions can be obtained, making PALM microscopy a key localization microscopy technique.
Signals from each fluorophore are still subject to the 300nm diffraction of light limit. However, because each has been activated separately, the centre of mass can be calculated accurately. The point spread function (PSF) is used to determine the location down to a resolution of 20nm. By mapping each of the more accurately defined points together, a complete super-resolution microscopy image can be compiled.
PALM microscopy can be combined with TIRF microscopy or HILO microscopy to improve the signal to noise ratio of thinner sample. This opens up possibilities for researchers studying membrane bound protein complexes or smaller vesicles.
PALM microscopy and STORM microscopy both use stochastic activation of individual fluorophores to resolve spatial details by identifying precise positions of single molecules. Three commonly used types of fluorophores are:
Photoactivatable Fluorophores: Today most common are photoactivatable proteins, eg: PAmCherry, PA-GFP, which emit light upon activation with UV.
Photoconvertible Fluorophores: Fluorescent proteins which change their emission spectrum upon activation with UV light (eg. mEOS proteins).
Photoswitchable Fluorophores: Typically chemical dyes (e.g. Alexa Fluor 647, DyLight555) which can switch between dark, non-fluorescent and bright, fluorescent states repeatedly.
Fluorophores are constantly evolving. New dyes are often photoactivated at higher wavelengths which are more favorable for live samples. Some fluorophores are now available that are both photoswitchable and photoactivatable, providing further options for imaging.
PALM microscopes can be used for fixed sample localization microscopy to describe the intricate details of analysed samples with resolutions reaching 20nm. Using photoactivatable or photoconvertible proteins enables counting of molecules, for example, in complexes and clusters.
Another application is single particle tracking in live cells. By stochastically activating individual fluorophores we are able to track movement of molecules over time, calculate diffusion coefficients and describe particle behavior.
As a PALM microscope, the Nanoimager's extraordinary sensitivity, stability and precision make it an excellent tool for precise measurements, using STORM and PALM microscopy. It is capable of imaging in four colors with two colors running simultaneously, allowing tracking of two distinct fluorophore species at the same time. This allows single molecule localizations and tracks registered in one channel to be relating to cellular markers in the second channel.
Lasers are digitally controlled providing accurate repeatable results and with a sCMOS camera, images can be generated down to 20nm resolution. It is a robust, unique design and so can be used in any lab location, not just optical tables in dark rooms. The design naturally dampens vibrations so clean imaging at high resolution becomes possible. The Nanoimager is versatile so it should not be considered as just a PALM microscope. It can also do dSTORM, single particle tracking, smFRET and SIM microscopy.
The Nanoimager has been used as a PALM microscope to study RNA polymerase tracking in E. coli, tracking of FtsK in Mycobacterium smegmatis and membrane complexes in bacteria.
Download our table comparing the options on the Nanoimager
Our team of scientists are waiting to help with your questions